Indoor Audible Alarms
Single Point Float Switches
Multi-Point Float Switches
Multi-Point Liquid Level Float Switches |
Ktech Model FS301-1
Model FS301-1 Multi-Point Liquid Level Float Switches
FS301-1 multi-point liquid level float switch set-point distances are measured from the underside of the 2 NPT bushing to the center of the float at the level where the switch is activated. Switch set-point distances are not field adjustable. They are factory-set as specified on the purchase order. The contact arrangement may be changed from normally open (NO) to normally closed (NC) and vice versa. Turn over the float and return the stop collars to their original positions. If the contact arrangement is altered, use an ohmmeter to test that the switch operates correctly. Screw the float switch into the 2 NPT opening on top of the tank. The conductors are colour coded to identify each switch. The switch must be connected in series with a load.
Model FS301-1 Construction:
DO NOT CONNECT THE SWITCH DIRECTLY ACROSS THE POWER SUPPLY.
The voltage / current product must not exceed the power rating.
Model FS301-1 Specifications
Manufacturer Of Float Switches, Vacuum Switches, Alarms And Pump Controllers.