All Myron L handheld instruments are factory calibrated with NIST traceable Standard Solutions having specific conductivity/ppm values. MyronL Company Standard Solutions are made under strictly controlled conditions using reagent grade salts. These salts are mixed with deionized water having a resistivity of at least 5 megohms-cm purity.
Myron L Company Standard Solutions have an accuracy of ±1% based on values published in the International Critical Tables and traceable to the National Institute of Standards and Technology.
The conductivity Standard Solutions and pH Buffers listed below are used for factory calibration. Regular use of these solutions is recommended to ensure specified instrument accuracy. Frequency of conductivity recalibration depends upon use, but once every month should be sufficient for an instrument used daily. pH models, depending upon use, should be recalibrated with pH 7 Buffer every 1-2 weeks, and checked with pH 4 and/or 10 Buffers at similar intervals. pH Sensor Storage Solution is recommended for keeping the pH sensor wet. Myron L solutions are available in quart/1 ltr., gallon/3,8 ltr. and 2 oz./59 ml plastic bottles, ready to use. (Custom value standards only offered in gallon/3,8 ltr. size bottles.)
Note: RE-10 Range Extenders are usually calibrated with either 442-15,000 or 442- 30,000 Standard Solution.
Conductivity instruments are a convenient way to determine the parts per million of total dissolved solids (ppm/TDS) in boilers, cooling towers, reverse osmosis systems, etc. Although the International Unit (Sl) of measuring conductivity is the microsiemens/ cm (also known as micromhos/cm), a direct reading in ppm/TDS is sometimes preferred.
The table below shows the Conductivity/TDS Equivalents for various Myron L Standard Solutions.
<< Back To Main Index >>
Search This Site
(Search by Product Type i.e. Level or Brand Name i.e. Walchem)