CAL-C110 pH and ORP Simulator
Sensorex - pH and ORP Checkers/Simulators
pH and ORP measurement loops can be difficult to troubleshoot unless you have the right tools. Our CAL-C110 simulator mimics the signal of a perfect pH or ORP sensor. No need to throw away good sensors when the issue may be the interconnection cables or pH/ORP instrument. The simulator can easily pay for itself the very first time you consider disposing of a perfectly good sensor. Allows the user to identify the sensor, cable, meter, preamplifier, or controller as the cause of the problem.
How the Simulator Can Help
If the controller displays the correct value, the electrode is bad, and needs to be replaced.
If there is still an error on the display, another system component is at fault, and the checker will help you to determine which one is bad. The simulator may be connected directly to the preamplifier, and the signal verified. if this solved the problem, the cable between the electrode and the preamplifier may be the culprit.
If the simulator is connected to the cables leaving the pre amp and the display is normal, the preamplifier needs to be checked. Maybe a loose cable connection on the terminal strip, a bad cable, or a bad preamplifier. if the simulator is connected directly to the controller, and the display still shows error, it may be time to service the controller.
The simulator costs less than most pH electrodes. It can easily pay for itself the very first time you consider disposing of a perfectly good pH electrode. You will be able to eliminate the electrode, cable, meter, preamplifier, or controller as the cause of the problem. You may also check under high impedance condition, a most useful option with our checkers.
Search This Site
(Search by Product Type i.e. Level or Brand Name i.e. Walchem)